- Material: ABS Dental Irrigator
- Model Number: Oral Irrigator for Teeth Cleaning
- Size: 22CM*4.5CM*2.8CM
- Item Type: Dental Flosser
- Name: Oral Irrigator
- Function: Tooth whitening
- Item name: water jet
- Power method: No batteries dental flosser
- Age Group: All
- Type: Portable Dental SPA Oral Irrigator for Teeth Cleaning
- Teeth Whitening: Oral Irrigation
- Instructions: Fill with water or mouthwash
- Dental Spa: Irrigator Oral
Remove debris and food residue from in between teeth,Preventing the formation of periodontitis
DEEP CLEAN FOR TEETH-A unique combination of water pressure and pulsations cleans deep between teeth and below the gumline where regular flossing can't reach
PORTABLE-easy to carry, replace toothbrush as travel products.Note:Our products are manual, no battery.
Adjust the normal mode according to the needs of different oral cavity clean, or slight flushing mode
EASY TO USE AND ADJUSTABLE-Provides and easy and effective way to support proper teeth and gum health at home
How to use portable oral irrigator?
1.Fill tank with water to max water level ''FULL'' not higher.
Notice:If you deside,you could add 'mouse rinse' into water tank.Use tap water or warm water(but water temperature should not exceed 40°)
2.Use bottom pump handle – push and pull about eight times(shown right figure)
3.Place the nozzle into the mouth
4.Use your thumb to press on the switch button to turn water jet on and off
Note:Press the switch intermittently for best results as you move along the teeth and gums.
5.Use it over the sink,keep lips slightly open to prevent water from splashing but to allow the water to drip into the sink
6.Keep the nozzle at about 45°along the teeth/gums direction and move it along as the jet is activated.This will clean food debris between teeth and gums.Do this in short bursts flushing until all the areas between the teeth and gums are cleaned
Not to be used for children under 6 yeara old.parent supervision required for children under 12 years old.
1. Individual product pressure is not enough, please hold the nozzle, press the button several times to loosen the nozzle, and then press the button .
2. Our products are ultrasonic splicing, can not be removed into two parts to clean, if you want to clean, the angle should not exceed 50 degrees. If conditions permit, you can use disinfection cabinet disinfection products.